The Action VIs – initiate or terminate test and measurement operations, such as arming the trigger system or generating a stimulus. Sep 2, However, when trying to run the plug-and-play examples, this error occurs “The ID Query failed. Back to Top 2. Of course Tulip was ticked for that to happen:
|Date Added:||12 February 2004|
|File Size:||66.1 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Sep 1, 6. Is this some sort of bug? Lastly, you don’t need to typecast the raw data into an I32 then typecast to SGL real.
You commanded the Agilent to send data in binary block format: Yes, my password is: There is a lot of good information easily found on the internet. Message 7 of LabVIEW instrument drivers simplify instrument control and reduce test program development labvoew by agllent the need for you to learn the complex, low-level programming commands for each instrument.
Hi, I am new to Labview so can you tell me how to uncheck the ID query please?
Back to Top 2. Most instrument drivers have menu palettes that include the following components. Generally, you need to call the Close VI only once at the end of an application or when you finish communication with an instrument. Each VI corresponds to a programmatic operation, such as configuring, reading from, writing to, and triggering an instrument. According to the description of the said driver, it is compatible with Agilent A.
How to Use an Instrument Driver in LabVIEW Tutorial – National Instruments
Then when I go to to run this driver, I see the next: Message 3 of 4. Sep 1, 2. Join Physics Forums Today! I did that and still had the same thing.
Agilent 8114 Signal Generator
As for the driver compatibility you will have to check the manuals to see if they use the same commands. Have something to add?
It would be extremely useful to show how to modify portions of instrument drivers for other use. The Action VIs – initiate or terminate test and measurement operations, such as arming the trigger system or generating a stimulus.
Timeout expired before operation completed. Enabling autozero optimizes the instrument for accuracy, while disabling optimizes for speed.
Most Active Software Boards: This would explain your missing data. What should I do if this is really the case?
Last edited by a moderator: Numerous Configuration VIs can exist, depending on the particular instrument. However, LabVIEW Plug and Play instrument drivers are distributed with their block diagram source code, so you can customize them for a specific application.
How to Use an Instrument Driver in LabVIEW Tutorial
We can quickly write the code for a trinket if you specify the frequency sweep parameters. Message 6 of If you are sure that you have selected the correct instrument and it is responding, try disabling the ID Query Error occurred at Agilent The Configuration VIs- a collection of software routines that configure the instrument xgilent perform the operation you ahilent.
I looked in the manual at the frequency, period setting functions and the index. Just wire a False boolean constant to the ID Query terminal of the initialize vi to bypass this check.