I’ve tried some more: It may not be immediately obvious which interface is best for control purposes. Details of chemical analyses – Proceedings of the Royal Society B. Message 2 of Even more I don’t have the aglent keysight names conflicting, I still call them HP.
|Date Added:||19 September 2016|
|File Size:||61.86 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Instrument Control (GPIB, Serial, VISA, IVI)
Please restart your computer and contact National Instruments if the problem persists. Click a thumbnail to view larger image: Hi, I apologize for my english, it is not my native language.
I clicked on “View Attributes” and ayilent the values: I suspect something went wrong with that. Here are several different scenarios: Stock Quick Agileht. However, there are still some programs that require the use of SICL. The names are confusing, thats right.
Best regards SteKo Solved! Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type. Documents Flashcards Grammar checker. In other words, socket communications may appear to work, but sooner or later will 11 cause system failures. This is optional, but it is a good way to check the connection. I already checked the tulip in MAX.
Examples of instruments that can communicate through sockets are the EA and EA mentioned above port The firmware server requires synchronous communications, where handshaking takes place to verify that the last command has been received and it is OK to send the next one. I think the conflict went away. I always used an ni gpib card so I did not have this problem.
Three alternatives allow the same functionality without some of these limitations.
Sampling O-Ring Kit – for Tulip and EMT Torches, , Thermo iCap
Therefore, I use the RS of the computer on the Mainboard. Does it appear under this category in Device Manager? Message 4 of This will allow interfacing GPIB devices to most newer computers without having to open them up to install interface cards. Message 4 of 5.
It worked without tulip so the adapter is working I guess. Message 9 of Finally, configure tullp desired interfaces. We appreciate your interest in using our InstraView website! In addition, the series and series of logic analyzers communicate through port Configure desired interfaces after refreshing or restarting MAX.
You must install revision 2. Any other way of communicating with the firmware server can cause it to crash. Note that the IP address must be changed to fit the situation. I have an agilent a instrument. I have done some more research: Reboot your machine, and tukip the issue goes away as it did for me.